My Upgrade Experience
I followed the instructions for the upgrade and for the most part, it went well. I had one glitch where an error code indicated that my pump was communicating with another device. I restarted the upgrade and everything went fine. I had not been using my Dexcom receiver and I was assured by other X2 users on Facebook that I could leave the G5 app running on my iPhone and Apple Watch. (Note: the Dexcom G5 transmitter can only communicate with one receiver and one smart device. You cannot use the X2 and the G5 receiver at the same time.)
For the first hour I received an unbroken every-5 minute tracing of my Dexcom readings. After that it went haywire and I got only 2 readings in the next hour. I continued to get multiple Out of Range alerts as the day went on.
When you use a tubed pump, your pump is never far away from your CGM transmitter. My Tandem pump was in a pocket. My Dexcom transmitter was on my arm. My iPhone was on the table, in a pocket, in the kitchen, in my purse and not missing a dot. The pump was struggling with 12 inches.
I called Tandem twice over the next couple of hours and with a little troubleshooting it was apparent that I shouldn’t be having such problems. As always, the Tandem reps were helpful and the second rep indicated that many similar problems had been solved by a new transmitter. My current transmitter has been in use for a month and seemingly fine, but I wasn’t going to argue with trying a new transmitter.
I received the new transmitter on Saturday and will begin using it with my next sensor.
Meanwhile I have begun to get better communication with my Dex sensor, but it is not because the pump is better at picking up the signal. It is because I am making changes. I initially kept my pump in my left front pocket with the sensor on my right arm. I am now wearing the pump clipped to my waistband on the right side. I still lose signal when I sit in my normal “easy chair” and my arm is against the back cushion. (My phone does not lose signal in this scenario.) But I am getting better at moving to the right side of the chair and keeping my arm on the armrest so that the transmitter is not obstructed. I prefer my pump in my pocket, but I can get used to the waistband.
Basically I am being flexible to make things work.
Somehow every change I make impacts something else. Wearing the pump on my waistband has brought back the problem that the clip on the new case is neither tight
Second hack. I am back to using a Nite Ize Hip Clip that is attached directly to the pump. When I used this clip a few months back, it also tended to fall off my waistband but not as
I am not done yet with figuring out a case hack and think it will involve a Nite Ize clip attached to the Tandem case and a small piece of Velcro. Or maybe just Velcro attached to the case using the Tandem clip. Although the Hip Clip attached directly to the pump is working OK, I prefer the protection of a case. My false occlusion alarms were eliminated by using a case and I am hesitant to go without one. I’ll definitely write another blogpost once I decide on an ultimate fix.
I like having my G5 information on my pump and I will continue to be as flexible as possible to make it work. Although there is some frustration that my relationship with the X2 is a bit temperamental—first due to occlusion alarms and now to CGM reception—I am still very happy with the pump. In general if I go back and look at the various pumps and CGMs that I have used since 2005, they have all required me to learn, change, and be flexible to ensure success.
Now I get to the nitty-gritty of this newest update. What are the Medicare repercussions??? I’m flexible, but I can’t say the same thing for Medicare.
I have discussed more than once the stupidity of the Medicare ban on smartphone use for Dexcom G5 users. I’m not going there today. However, I have been hoping that because the Tandem X2 is durable medical equipment (DME) and not a smartphone that I will be able to use it as my CGM receiver instead of the Dexcom receiver.
When I read the Noridian Medicare coding and coverage document released in March, I do not believe that the X2 violates the guidelines:
“Coverage of the CGM system supply allowance is limited to those therapeutic CGM systems where the beneficiary ONLY uses a receiver classified as DME to display glucose data. If a beneficiary uses a non-DME device (smart phone, tablet, etc.) as the display device, either separately or in combination with a receiver classified as DME, the supply allowance is non-covered by Medicare.”
Unfortunately I have heard from several sources that Dexcom is instructing Medicare customers that current Medicare instructions mandate use of the Dexcom receiver. I suppose justification for that comes from the first sentence of the Noridian document: “The Dexcom G5® Mobile CGM System is currently the only FDA-approved device with a “non-adjunctive” indication.” Although my Tandem pump works with the G5 Mobile CGM System, it is technically not part of the system according to Medicare. Some Tandem pumpers have also been told by Tandem that the X2 is currently not approved by Medicare to be used as a Dexcom G5 receiver.
I continue to be optimistic that Medicare regulations regarding the Dexcom G5 will be changed to allow use of a smartphone and use of the Tandem X2. I have no predictions for a timetable for those changes. In the short run I am still using Dexcom supplies purchased before Medicare and continue to use my iPhone, Apple Watch, and Tandem X2 pump. In the long run I will follow Medicare regulations because CGM coverage is too valuable to risk losing that coverage.
As always with diabetes, I will be flexible.